Sinumerik Alarm 61302 Channel %1 block %2 measuring probe collision
%1 = Channel number
%2 = Block number, label
The measuring probe collided with an obstacle when being positioned.
– Interpreter stop
– NC Start disable in this channel.
– Interface signals are set.
– Alarm display.
• Check spigot diameter (may be too small) check measuring distance (may be to long)
Clear alarm with the RESET key. Restart part program
- Siemens Diagnostics Guide by Siemens